- Open Access
Hyperspectral hybrid method classification for detecting altered mucosa of the human larynx
© Martin et al.; licensee BioMed Central Ltd. 2012
Received: 19 March 2012
Accepted: 29 May 2012
Published: 21 June 2012
In the field of earth observation, hyperspectral detector systems allow precise target detections of surface components from remote sensing platforms. This enables specific land covers to be identified without the need to physically travel to the areas examined. In the medical field, efforts are underway to develop optical technologies that detect altering tissue surfaces without the necessity to perform an excisional biopsy. With the establishment of expedient classification procedures, hyperspectral imaging may provide a non-invasive diagnostic method that allows determination of pathological tissue with high reliability. In this study, we examined the performance of a hyperspectral hybrid method classification for the automatic detection of altered mucosa of the human larynx.
Materials and methods
Hyperspectral Imaging was performed in vivo and 30 bands from 390 to 680 nm for 5 cases of laryngeal disorders (2x hemorrhagic polyp, 3x leukoplakia) were obtained. Image stacks were processed with unsupervised clustering (linear spectral unmixing), spectral signatures were extracted from unlabeled cluster maps and subsequently applied as end-members for supervised classification (spectral angle mapper) of further medical cases with identical diagnosis.
Linear spectral unmixing clearly highlighted altered mucosa as single spectral clusters in all cases. Matching classes were identified, and extracted spectral signatures could readily be applied for supervised classifications. Automatic target detection performed well, as the considered classes showed notable correspondence with pathological tissue locations.
Using hyperspectral classification procedures derived from remote sensing applications for diagnostic purposes can create concrete benefits for the medical field. The approach shows that it would be rewarding to collect spectral signatures from histologically different lesions of laryngeal disorders in order to build up a spectral library and to prospectively allow non-invasive optical biopsies.
Spectral examinations using remote sensing data have proven to give insights on the distribution, structure and condition of different land covers . Hyperspectral imaging improves accuracy and significance of such examinations  by providing the analyst with in-depth spectral data from the visible to the infrared bandwidth , thus overcoming the limitations of conventional imaging techniques. In practice, hyperspectral detector systems such as AVIRIS  and HYDICE  record radiation reflected from the ground in 200+ spectral channels from 0.4 to 2.5 μm simultaneously on each spot of a large array of spatial positions . Advances made in recent years have also led to vast extensions of the possibilities for computational processing. Among the various methodologies for image analysis, such as anomaly detection  and material identification , the procedure of target detection reveals great advantage for pointed surface identifications . Here the recorded scene is being scanned for a certain reflectance pattern (‘target signature’) that is known beforehand and that represents the spectral characteristics of a target surface in question. Thus, statements can be made on the appearance and distribution of certain vegetation [10–12], current levels of water stress  and pest infestation [14, 15], morphology, soil topography  and other features.
In the medical field, much effort is spent on the development of reliable diagnostic techniques detecting pathological changes on tissue surfaces [17–19]. Thereby, diagnostic procedures are preferably as minimal invasive as possible to avoid interference with the patient’s tissue. Among the newer techniques, spectral optical imaging proves very promising for future applications [20–24]. Hyperspectral imaging in particular may provide a procedure that is less invasive than conventional diagnostic techniques and has potential for improved detection of abnormalities and pathological surface alterations [25, 26]. Even though hyperspectral biomedical imaging has been well established on the technical side [27, 28], corresponding applications often lack expedient classification procedures for adequate image analysis. The question, therefore, is whether additional benefits may be attained for the medical field when existing procedures taken from remote sensing applications are used to meet biomedical demands.
Both supervised and unsupervised classification procedures have been applied to hyperspectral surface detections of remotely sensed data [29, 30]. In supervised learning, classification results highly depend on the number and quality of labeled samples . Thus, errors can be minimized when the data is represented by a sufficient number of classes and class signatures are well separable in numerical feature space . In unsupervised learning, spectral clusters and target areas are not necessarily coherent, but depend on the signature’s characteristics. Thus, final classes may not represent the actual regions of interest, even though unsupervised methods show insensitivity towards the dimensionality of spectral input . An approach to meet both requirements - adequate end-member classes and potentially meaningful target assignments - is the so called Hybrid Method, which combines both supervised and unsupervised classification in a single procedure .
In this study, target detections using the hybrid method classification are performed on hyperspectral images of mucosal surface alterations of the human larynx. As no reference spectra for laryngeal disorders have yet been collected, the approach extracts spectral information from unsupervised clustering for subsequent supervised classification. Thus, there is no need to generate adequate training sites in every image scene, which would require the expertise of clinical experts with longstanding experience. The study is aimed at examining the principle applicability and performance of signature-based classifications to an automated detection of laryngeal disorders. In this way, the investigation is designed as a feasibility study that proposes a first attempt to assess the qualification of hyperspectral signatures for diagnostic detections. We hypothesize that pathological subregions of human vocal cords can be identified by means of unsupervised clustering, and that extracted hyperspectral signatures from unlabeled cluster maps can be applied to automatic target detections in second medical cases with identical diagnosis. Thus, analytical tools derived from remote sensing demands can be applied to demands of diagnostic recordings of suspicious tissue surfaces, providing an investigative attempt to treat hyperspectral data in medical studies and enabling further research on non-touch optical biopsies.
Materials and methods
Five patients (cases #1-#5) with two histological findings (2x hemorrhagic polyp, 3x leukoplakia) were included in our study. Microlaryngoscopies were scheduled by arrangement and informed consent was obtained. Hyperspectral recording was performed in vivo. Endoscopies were realized using two different operation microscopes (OPMI Pentero and Vario, Carl Zeiss Meditec AG) under general anesthesia in jet-ventilation, as previously described in . In order to generate monochromatic images at a consistent array of 10 nm wavelength steps, a Polychrome V (TILL Photonics) was synchronously triggered with a monochromatic high-resolution camera (AxioCam HRm, Zeiss MicroImaging). The camera and the Polychrome V light source were connected to both microscope types using a C-mount thread module and integrated optical fibres. Other than that, no additional changes to the standard operation setup were made, i.e. no further sampling or excisional biopsy was conducted. Subsequently, the resulting image stacks were cropped by narrowing down the field of view to the relevant parts of the vocal cords. Here the aim was to generate images of homogeneous illumination that show pathological tissue changes on the vocal cords in a purely biological environment. This is due to the fact that illumination was not homogeneous in all recorded stacks and that technical parts of the endoscope often protruded into marginal areas of the images.
where f = proportion of the nth surface component (n = 1,…, N), μ = expected spectral signal from the nth surface component, and N = total number of components . The applied clustering approach aimed to provide a hierarchical order of the ‘strongest’ spectral signatures present in the image scene. Therefore, a first pixel was identified whose reflection value was the highest among all pixels present. Following equation (1), this is the pixel whose sum of squares of all reflection values equaled the maximum numeric value of all pixels. The reflection values of that pixel in all channels were extracted and subsequently applied as an end-member signaturea for cluster 1. As it is an internal criterion of spectral clustering , the unsupervised clustering approach further aimed at providing this hierarchical order of spectral signatures with the highest possible separability in numerical feature spaceb, thus preparing signatures for end-member subsets with high intra-cluster similarity and low inter-cluster similarity . The second signature was therefore chosen by finding the pixel with reflection values in the respective channels that were most different from those of signature 1 in numerical terms. To this end, the reflection values in the respective channels were subtracted from those of the corresponding channels of signature 1. The resulting differences were squared and summed in order to identify the highest calculated value. The set of reflection values corresponding to the highest value described highest possible separability from signature 1 and thus was accepted as the end-member signature for cluster 2.
In our study, this procedure continued until a predefined number of 20 signatures was generated for all five image stacks. This relatively high number was chosen in light of the subsequent classification processes whose quality of data representation depends significantly on a sufficient number of spectral classes. Furthermore, the number resulted from our empirical experience. While clusters from 1 to 20 revealed at least one class with notable correlation to altering tissue locations, a cluster number of over 20 revealed clusters of increasingly useless information. In order to develop unlabeled cluster maps, posterior probabilities were calculated for each pixel to belong to one of the 20 classes. A comparison of probabilities yielded the maximum likelihood for each pixel’s affinity and classes were assigned accordingly. As the approach of the hybrid method classification was aimed at automatic target detections of pathological tissue expansions on human vocal cords, resulting clusters from unlabeled cluster maps were compared to abnormal surface expansions on the corresponding white light images with the support of clinical experts and operation protocols. This means that a visual interpretation by ENT (Ear-Nose-Throat) specialists and an inclusion of medical records was prompted to undertake a first assessment of the cluster locations. By highlighting the suspicious subregions of vocal cords as single spectral clusters, these clusters already gave an indication of the applicability of hyperspectral classifications to the applied mucosal surface recordings, even though the general qualification of the procedure has already been proven for previous recordings of that same area .
The final supervised classification aimed at automatic target detection was performed exemplarily for one diagnostic pair in each of the 2 histological findings. Therefore, unlabeled cluster maps of case #1 and #3 served as training sites to extract spectral signatures for all 20 classes from the respective 30 channels. The acquired end-members of case #1 were then applied to the final supervised classification of case #2 among the hemorrhagic polyps, and those of case #3 were applied to the final supervised classification of case #4 among the cases of leukoplakia. The spectral angle mapper algorithm was chosen for mapping spectral similarity between image spectra and cluster spectra in order to perform supervised classification. The algorithm is a well established similarity measure in hyperspectral classifications [42–44]; it treats reference and field spectra as vectors in n-dimensional feature space, with n corresponding to the number of bands and different vector positions representing differences in illumination for a target material. Similarity between spectra is then determined by calculating the resulting angle between regarded vectors. As the angle between both spectra remains constant for all examined positions , the approach offers the advantage of neglecting disturbances caused by light/shadow contrasts.
As the results of the automatic target detections in case #2 and #4 corresponded to a common two-group outcome (pathological/non-pathological), a reclassification of the 20 clusters to a binary decision map delivered the diagnostic expression for both detections. The value 1 (positive result) corresponded to the ‘pathological’ condition, while the value 0 (negative result) corresponded to the ‘normal’ condition. To perform a statistical validation, confusion matrices were then calculated as described in . In our case, designated pixels from detection outcomes were compared to reference masks that illustrated the known diagnosis by defining the exact location of the altered tissue in cases #2 and #4 (corresponding masks were created with help of ENT expert knowledge). For assessment of the diagnostic accuracy, the traditional measures of sensitivity and specificity were calculated. As the sensitivity of a test refers to its capability to identify positive results , it calculated our model's ability to correctly detect pixels of pathological tissue. Specificity, on the other hand, refers to a test’s capability to identify negative results and calculated our model's ability to correctly diagnose pixels of healthy tissue. In addition, the bias was calculated to quantify possible over- or underestimations in our detections. Since the bias is the total number of positively detected pixels divided by the total number of truly positive pixels according to the reference mask, an ideal bias has the value of 1. A bias > 1 indicates an overestimation, whereas a bias < 1 indicates an underestimation .
The hyperspectral imaging workstation was set up without technical problems. Both microscope types could readily be connected to the Polychrome V light source and the high resolution camera by applying the C-mount connector. The respiration stop during jet ventilation for movement reduction of the larynx (30 sec) posed no anesthesiological problem for any of the patients. The hyperspectral imaging procedure provided hyperspectral image stacks containing 30 spectral bands from 390 to 680 nm for all cases. Each image stack could be treated as a series of channels that were associated with identical pixel locations. This enabled further performance of hyperspectral routines, whereby the image stacks represented collections of various, evenly spaced wavelength bands recorded for one region of interest.
Confusion matrices with sensitivity/specificity analysis and calculated bias for the automatic target detections in case #2 and #4
Automatic target detection in case #2
Automatic target detection in case #4
Crucial differences exist between given circumstances in remote sensing applications of hyperspectral imaging and those of biomedical uses. Hyperspectral imaging in satellite observation utilizes the sun as light source, providing the target area with the entire spectrum of electromagnetic radiation. Spectral separation of the reflected energy for analysis is then performed during the recording procedure. In biomedical applications, crucial changes in the testing setup have to be made in order to facilitate adequate illumination of a biological region of interest. In the context of this study, the larynx is a straitened area of the human body with no natural sunlight interfering. These circumstances necessitate the need to implement refractors, monochromators or tunable filters to separate the light into single bands. Thus, illumination was performed exposing the larynx by means of a monochromatic light source. Also, no additional filter was desired between the mucosa and the camera, since illumination was limited to a confined number of reflecting photons in our test setup. Eventually, adequate conditions could be established by means of monochromatic illumination and synchronously triggered high-resolution imaging, allowing modern medicine to perform hyperspectral imaging and opening doors to potentially benefit from existing data processing techniques derived from the field of earth remote sensing.
In this study on the application of hyperspectral hybrid method classification to automatically detect laryngeal disorders, good results could be achieved throughout all utilized procedures. The results indicate that areas of hemorrhagic vocal cord polyps and present leukoplakia can be recognized as independent elements of hyperspectral image compositions using unsupervised image clustering. We can thus infer a considerable separability of pathological mucosa from environmental tissue with an unexpectedly high degree of accuracy. Because no additional input of reference spectra was applied, class matches were attained automatically. This also speaks for the clustering processes’ capability to diagnose abnormal tissue changes as being divergent from surrounding mucosa. No considerable difference could be perceived in the quality of recognition between the different histopathological groups. However, spectral profiles from clusters covering pathological tissue locations delivered notable differences. While in cases #1 and #2, profiles were still alike in their general course throughout all channels, similarities in shape among the cases #3, #4 and #5 could only be found for certain segments of the entire spectrum. In numerical terms, cases clearly differed in both histopathological groups. This particularly applied to the three cases of a present leukoplakia between 590 and 680 nm. In view of the region examined, these variabilities in spectral response may be ascribed to the specific manifestations of a disease that vary from patient to patient. Even though significant spectral similarities should be assumed for altered surface tissue with identical diagnoses, spectral responses will accordingly vary due to natural pathogenetic peculiarities. Further, differences in the anatomical environment of the larynx caused differences in illumination during hyperspectral recording in all cases.
Final supervised classifications unambiguously assigned pixels of second medical cases #2 and #4 to the signature classes from unlabeled cluster maps of initial cases #1 and #3. With special regard to the classes matching with local alterations in the initial stacks, results surprised by showing primarily those pixels being ascribed, which represented the core area of the second case’s own pathological expansion. Even though according classes also spread into further areas of the view frame, mistakenly assigning few locations of healthy tissue to the ‘pathological’ signatures, the centers of pathological expansions were identified precisely. Thus, only minor failures distort the result of an accurate and locally correct diagnosis in both cases as confirmed by statistical validation. Even though a notable difference could be assessed in the sensitivity of both detections, the training set of 2 patients is too small to draw conclusions on the methods’ qualification on different diseases yet. However, the results are particularly meaningful in view of the above mentioned spectral variability of the examined abnormality appearances. At that point, measuring spectral similarity between the target and the pixel spectra by means of the spectral angle mapper appeared to be a useful choice. It improved classification results by disregarding specific surface irregularities of the vocal cords that naturally led to inhomogeneous reflections in every patient. Consequently, it ascribed identical mucosa alterations in both cases to the same spectral cluster and thus, coped the problem of spectral variability of equal surface components notably well, which remains a major challenge for classifications and target detections in hyperspectral imagery.
These results show that the initially postulated hypotheses that pathologic subregions of human vocal cords can be identified by means of unsupervised clustering and that extracted hyperspectral signatures from unlabeled cluster maps can be applied in a practical context to automatic target detections in second medical cases of identical diagnosis can be confirmed. Furthermore, the overarching question of this study whether the medical field can profit from taking existing procedures from the field of remote sensing and applying them to biomedical demands, where imaging modalities are already able to deliver corresponding image stacks, can be answered affirmatively.
The insights attained on the performance and potential applicability of hyperspectral classifications for the automatic detection of laryngeal disorders may give an impetus for further research on hyperspectral classifications in medical usage. The underlying reason is that laryngeal disorders of the vocal cords cause specific changes in the surface structure of the covering epithelium. These alterations, often expressed by atypical modifications of color and texture, pave the way for spectral determinations using automatic target detections. This study was performed exemplarily for five image stacks featuring two different histopathological findings of benign alterations. As it is known that carcinoma of the vocal cords significantly contrast with their non-pathological environment likewise (albeit with a higher variability of spectral characteristics), similar results may be expected for malignant proliferations. More specifically, carcinoma of the human larynx should be equally identifiable by means of hyperspectral clustering procedures, while an establishment of automatic target detections may require a larger set of reference spectra, as well as proper calibrations of statistically suitable similarity measures, to compensate for spectral variability. However, this potential is particularly relevant for the progress of cancer diagnostics and shall be the subject of upcoming investigations.
Given a sufficient number of clinical cases, spectral signatures can be collected for histologically different lesions of laryngeal disorders in order to build up spectral libraries covering a wide variety of clinical findings. More complex classification procedures may then be applied that are specifically designed for use with library spectra and will improve detection accuracy by taking into account the severity and stage of a disease. Consequently, first attempts of clinical trials would be allowed, making an essential step towards the establishment of non-touch optical biopsy.
a The term end-member signature denotes the spectral response pattern of the pixel that defined the class metrics of a certain cluster for spectral unmixing.
b In this case, numerical feature space is an abstract space with a feature number equal to the number of spectral channels. Each signature is a defined position in 30-dimensional space, which enables numerical quantification of their separability.
This project was jointly realized by the Laboratory for Climatology and Remote Sensing at University of Marburg and the Department of Otorhinolaryngology, Head and Neck Surgery at University of Bonn. The authors thank Prof. Dr. Jörg Bendix and Prof. Dr. med. Dr. h.c. Friedrich Bootz for their trust and support. The help of Martin Schulz on the statistical analysis is highly acknowledged.
- Goetz AFH, Vane G, Solomon JE, Rock BN: Imaging Spectrometry for Earth Remote Sensing. Science. 1985, 228: 1147-1153. 10.1126/science.228.4704.1147.View ArticlePubMedGoogle Scholar
- Camps-Valls G, Bandos Marsheva TV, Zhou D: Semi-supervised Graph-based Hyperspectral Image Classification. Transactions on Geoscience and Remote Sensing. 2007, 45: 3044-3054.View ArticleGoogle Scholar
- Rajendran S, Rajamanickam M: Hyperspectral Remote Sensing – A Development and Perspective. Hyperspectral Remote Sensing and Spectral Signature Applications. Edited by: Rajendran S, Aravindan S, Jeyavel Rajakumar T, Sivakumar R, Murali Mohan KR. 2009, New India Publishing Agency, New Delhi, 1-9. 1Google Scholar
- Green RO, Eastwood ML, Sarture CM, Chrien TG, Aronsson M, Chippendale BJ, Faust JA, Pavri BE, Chovit CJ, Solis M, Olah MR: Imaging Spectroscopy and the Airborne Visible Infrared Imaging Spectrometer (AVIRIS). Remote Sens Environ. 1998, 65: 227-248. 10.1016/S0034-4257(98)00064-9.View ArticleGoogle Scholar
- Rickard LJ, Basedow RW, Zalewski EF, Silverglate PR, Landers M: HYDICE: An Airborne System for Hyperspectral Imaging. Proceedings of SPIE : 14 April 1993. Edited by: Vane G. 1937, SPIE Press, Orlando, 173-179.Google Scholar
- Plaza A, Benediktsson JA, Boardman JW, Brazile J, Bruzzone L, Camps-Valls G, Chanussot J, Fauvel M, Gamba P, Gualtieri A, Marconcini M, Tilton JC: Trianni G: Recent advances in techniques for hyperspectral image processing. Remote Sens Environ. 2009, 113: 110-122.View ArticleGoogle Scholar
- Stein DWJ, Beaven SG, Hoff LE, Winter EM, Schaum AP, Stocker AD: Anomaly detection from hyperspectral imagery. Signal Processing Magazine. 2002, 19: 58-69. 10.1109/79.974730.View ArticleGoogle Scholar
- Healey G, Slater D: Models and Methods for Automated Material Identification in Hyperspectral Imagery Acquired under Unknown Illumination and Atmospheric Conditions. Transactions on Geoscience and Remote Sensing. 1999, 37: 2706-2717. 10.1109/36.803418.View ArticleGoogle Scholar
- Caefer CE, Rotman SR, Silverman J, Yip PW: Algorithms for point target detection in hyperspectral imagery. Proceedings of SPIE. Edited by: Sylvia S. 2002, Shen. San Diego: SPIE Press, Seattle, 424-457. 4816Google Scholar
- Kalacska M, Bohlman S, Sanchez-Azofeifa GA, Castro-Esau K, Caello T: Hyperspectral discrimination of tropical dry forest lianas and trees: Comparative data reduction approaches at the leaf and canopy levels. Remote Sens Environ. 2007, 109: 406-415. 10.1016/j.rse.2007.01.012.View ArticleGoogle Scholar
- Gong P, Pu R, Yu B: Conifer species recognition: An exploratory analysis of in situ hyperspectral data. Remote Sens Environ. 1997, 62: 189-200. 10.1016/S0034-4257(97)00094-1.View ArticleGoogle Scholar
- Hamada Y, Stow DA, Coulter LL, Jafolla JC, Hendricks LW: Detecting Tamarisk species (Tamarix spp.) in riparian habitats of Southern California using high spatial resolution hyperspectral imagery. Remote Sens Environ. 2007, 109: 237-248. 10.1016/j.rse.2007.01.003.View ArticleGoogle Scholar
- Zarco-Tejada PJ, González-Dugo V, Berni JAJ: Fluorescence, temperature and narrow-band indices acquired from a UAV platform for water stress detection using a micro-hyperspectral imager and a thermal camera. Remote Sens Environ. 2012, 117: 322-337.View ArticleGoogle Scholar
- Prabhakar M, Prasad YG, Thirupathi M, Sreedevi G, Dharajothi B, Venkateswarlu B: Use of ground based hyperspectral remote sensing for detection of stress in cotton caused by leafhopper (Hemiptera: Cicadellidae). Computers and Electronics in Agriculture. 2011, 79: 189-198. 10.1016/j.compag.2011.09.012.View ArticleGoogle Scholar
- Glenn NF, Mundt JT, Weber KT, Prather TS, Lass LW, Pettingill J: Hyperspectral data processing for repeat detection of small infestations of leafy spurge. Remote Sens Environ. 2005, 95: 399-412. 10.1016/j.rse.2005.01.003.View ArticleGoogle Scholar
- Silvestri S, Marani M, Marani A: Hyperspectral remote sensing of salt marsh vegetation, morphology and soil topography. Physics and chemistry of the Earth. 2003, 28: 15-25. 10.1016/S1474-7065(03)00004-4.View ArticleGoogle Scholar
- Govindaraju SK, Ahmed HU, Sahu M, Emberton M: Tissue characterisation in prostate cancer using a novel ultrasound approach. British Journal of Medical and Surgical Urology. 2008, 1: 98-106. 10.1016/j.bjmsu.2008.09.003.View ArticleGoogle Scholar
- Schmidt LH, Biesterfeld S, Kümmel A, Faldum A, Sebastian M, Taube C, Buhl R, Wiewrodt R: Tissue Microarrays are Reliable Tools for the Clinicopathological Characterization of Lung Cancer Tissue. Anticancer Res. 2009, 29: 201-210.PubMedGoogle Scholar
- Bige Y, Hanfeng Z, Rong W: Analysis of microstructural alterations of normal and pathological breast tissue in vivo using the AR cepstrum. Ultrasonics. 2006, 44: 211-215. 10.1016/j.ultras.2005.11.001.View ArticlePubMedGoogle Scholar
- Johansson A, Kromer K, Sroka R, Stepp H: Clinical optical diagnostics – Status and Perspectives. Medical Laser Application. 2008, 23: 155-174. 10.1016/j.mla.2008.08.002.View ArticleGoogle Scholar
- Balas C: Review of biomedical optical imaging - a powerful, non-invasive, non-ionizing technology for improving in vivo diagnosis. Measurement science and technology. 2009, 20: 1-12.View ArticleGoogle Scholar
- Charvet I, Thueler P, Vermeulen B, Saint-Ghislain M, Biton C, Jacquet J, Bevilacqua F, Depeursinge C, Meda P: A new optical method for the non-invasive detection of minimal tissue alterations. Phys Med Biol. 2002, 47: 2095-2108. 10.1088/0031-9155/47/12/307.View ArticlePubMedGoogle Scholar
- Farkas DL, Becker D: Applications of spectral imaging: detection and analysis of human melanoma and its precursors. Pigment Cell Res. 2001, 14: 2-8. 10.1034/j.1600-0749.2001.140102.x.View ArticlePubMedGoogle Scholar
- Chung S, Cerussi A, Mehta R, Hsiang D, Tromberg B: Non-invasive detection and monitoring of tumor pathological grade during neoadjuvant chemotherapy by measuring tissue water state using diffuse optical spectroscopic imaging. Cancer Res. 2009, 69 (Suppl 2): 803-View ArticleGoogle Scholar
- Dicker DT, Lerner J, Van Belle P, Barth SF, Guerry D, Herlyn M, Elder DE, El-Deiry WS: Differentiation of normal skin and melanoma using high resolution hyperspectral imaging. Cancer Biol Ther. 2006, 5: 1033-1038. 10.4161/cbt.5.8.3261.View ArticlePubMedGoogle Scholar
- Grosberg LE, Radosevich AJ, Asfaha S, Wang TC, Hillman EMC: Spectral Characterization and Unmixing of Intrinsic Contrast in Intact Normal and Diseased Gastric Tissues Using Hyperspectral Two-Photon Microscopy. PLoS One. 2011, 6: e19925-10.1371/journal.pone.0019925.PubMed CentralView ArticlePubMedGoogle Scholar
- Vo-Dinh T, Stokes DL, Wabuyele MB, Martin ME, Song JM, Jagannathan R, Michaud E, Lee RJ, Pan XG: A hyperspectral imaging system for in vivo optical diagnosis. Engineering in Medicine and Biology Magazine. 2004, 23: 40-49. 10.1109/MEMB.2004.1360407.View ArticlePubMedGoogle Scholar
- Liu HY, Li QL, Wang YTi, Liu JG, Xue YQ: Molecular Hyperspectral Imaging (MHSI) System and Application in Biochemical Medicine. Spectroscopy and Spectral Analysis. 2011, 31: 2593-2597.PubMedGoogle Scholar
- Averbuch A, Zheludev M: Two Linear Unmixing Algorithms to Recognize Targets using Supervised Classification and Orthogonal Rotation in Airborne Hyperspectral Images. Remote Sensing. 2012, 4: 532-560. 10.3390/rs4020532.View ArticleGoogle Scholar
- Chiang S-S, Chang C-I, Ginsberg IW: Unsupervised target detection in hyperspectral images using projection pursuit. Transactions on Geoscience and Remote Sensing. 2001, 39: 1380-1391. 10.1109/36.934071.View ArticleGoogle Scholar
- Kang JM, Guo L: A Semi-Supervised Approach to Image Classification in an Urban Environment. [http://www.spatial.cs.umn.edu/Courses/Fall07/8715/sample_project_report_3.pdf]
- Schowengerdt RA: Remote sensing, models and methods for image processing. 1997, Academic, OxfordGoogle Scholar
- Gerstner AOH, Laffers W, Bootz F, Farkas DL, Martin R, Bendix J, Thies B: Hyperspectral imaging of mucosal surfaces in patients. Journal of Biophotonics. 2012, 5: 255-262. 10.1002/jbio.201100081.View ArticlePubMedGoogle Scholar
- Wang L, Uchida S: Use of Linear Spectral Mixture Model to estimate Rice Planted Area based on MODIS data. Rice Science. 2008, 15: 131-136. 10.1016/S1672-6308(08)60031-1.View ArticleGoogle Scholar
- Miao X, Gong P, Swope S, Pu R, Carruthers R, Anderson GL, Heaton JS, Tracy CR: Estimation of yellow starthistle abundance through CASI-2 hyperspectral imagery using linear spectral mixture models. Remote Sensing of Environment. 2006, 101: 329-341. 10.1016/j.rse.2006.01.006.View ArticleGoogle Scholar
- Heinz DC: C-I Chang: Fully Constrained Least Squares Linear Spectral Mixture Analysis Method for Material Quantification in Hyperspectral Imagery. IEEE Transactions on Geoscience and Remote Sensing. 2001, 39: 529-545. 10.1109/36.911111.View ArticleGoogle Scholar
- Roberts DA, Gardner M, Church R, Ustin S, Scheer G, Green RO: Mapping chaparral in the Santa Monica Mountains using multiple endmember spectral mixture model. Remote Sensing of Environment. 1998, 65: 267-279. 10.1016/S0034-4257(98)00037-6.View ArticleGoogle Scholar
- Foody GM, Cox DP: Sub-pixel land cover composition estimation using a linear mixture model and fuzzy membership functions. International Journal of Remote Sensing. 1994, 15: 619-631. 10.1080/01431169408954100.View ArticleGoogle Scholar
- Settle JJ, Drake NA: Linear mixing and the estimation of ground cover proportions. International Journal of Remote Sensing. 1993, 14: 1159-1177. 10.1080/01431169308904402.View ArticleGoogle Scholar
- Manning CD, Raghavan P, Schütze H: Introduction to Information Retrieval. 2008, Cambridge University Press, New YorkView ArticleGoogle Scholar
- Bach FR, Jordan MI: Learning Spectral Clustering, With Application To Speech Separation. Journal of Machine Learning Research. 2006, 7: 1963-2001.Google Scholar
- Yang C, Everitt JH, Bradford JM: Yield estimation in hyperspectral imagery using Spectral angle mapper (SAM). Transactions of the ASABE. 2008, 51: 729-737.View ArticleGoogle Scholar
- Garcia-Allende PB, Conde OM, Mirapeix J, Cubillas AM, Lopez-Higuera JM: Data processing method applying principal component analysis and spectral angle mapper for imaging spectroscopic sensors. IEEE Sensors Journal. 2008, 8: 1310-1316.View ArticleGoogle Scholar
- Hillnhütter C, Mahlein A-K, Sikora RA, Oerke EC: Remote Sensing to detect plant stress induced by Heterodera schachtii and Rhizoctonia solani in sugar beet fields. Field Crops Research. 2011, 122: 70-77. 10.1016/j.fcr.2011.02.007.View ArticleGoogle Scholar
- Kruse FA, Lefkoff AB, Boardman JW, Heidebrecht KB, Shapiro AT, Barloon PJ, Goetz AFH: The Spectral Image Processing System (SIPS) – Interactive Visualization and Analysis of Imaging Spectrometer Data. Remote Sensing of Environment. 1993, 44: 145-163. 10.1016/0034-4257(93)90013-N.View ArticleGoogle Scholar
- Kohavi R, Provost F: Glossary of Terms. Editorial for the special issue on applications machine learning and the knowledge discovery process. Machine Learning. 1998, 30: 271-274.View ArticleGoogle Scholar
- Altman DG, Bland JM: Statistic Notes: Diagnostic tests 1: sensitivity and specificity. British Medical Journal. 1994, 308: 1552-10.1136/bmj.308.6943.1552.PubMed CentralView ArticlePubMedGoogle Scholar
- Stanski HR, Wilson LJ, Burrows WR: Survey of common verification methods in meteorology. Atmospheric Environment Service. 1989, Research Report Nr. 89-5: 1-81.Google Scholar
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.